- Dedicated pumping system for the analysis - Total flow
- Pumping system placed on a cart for easy maintenance
- Dedicated Electric and Pneumatic cabinet
- Ventilation and Sound proofing of the pumping system
- Totally customizable HMI interface
The proposed system provides three fixed testing stations, each of which can accommodate nr. 5 blood filters of different sizes manually re-arranging the same positions. Aligned to the testing stations there will be three vacuum chambers, one per position, including the poses by horizontal movement. On the other side of the test stations and on the opposite side to the vacuum chambers, there will be three unloading stations after the blood filter test, each post will be combined with the relative test post. The blood filters are taken by special suction cups assembled on equipment combined with a precision pneumatic cylinder which will in turn be assembled on a linear axis that will perform the loading/unloading of the filters blood samples on the three test stations. The discharge of the blood filters after the test will be performed by three linear axes, one for each testing station, on which will be assembled three precision pneumatic cylinders and the suction cups for the removal of the filters themselves. Behind the vacuum chambers will be placed the pumping unit, unique for all three chambers, housed on a support frame inside a soundproofed and ventilated cabin. The cabinets containing the electrical panel, the analysis system, and the three pneumatic distribution panels for the process gas will be solid to the pump cabin. The HMI interface will consist of a color touch-screen monitor and indicator lights for the operator/maintenance technician.
Productivity: 5 Filters every 13'' seconds with n° 5 parts tested per chamber - Reject threshold: higher than 3,0E-5 mbar*l/sec (helium 100% ) - Test Pressure (max): 300 mbar